To give the test engineer the ability to check the fixture calibration data.
Checking and Saving the Fixture Calibration Data:
1. Inspect calibration data before saving it to the fixture. Begin by high lighting the fixture you are calibrating. Using a right mouse button click select Calibration Inspect from the menu.
2. From the list of Fixture RF paths that will appear, high light the path that is to be inspected ( Figure 1). Use a right mouse button click and choose View / Rectangular ( Figure 2 )as selections. A plot of the calibration data will appear.
3. Use the graph format choices to set up the graph with proper units and references for viewing. A sample set of data is
provided in Figures 3-6.
4. Once the fixture calibration data has been check it can be saved. To do this high light the activated fixture and with a right hand mouse button click bring up the selection menu. Choose Save Calibration to save the data just taken to the fixture.