Normal Support procedures require that the TIM be shipped back to RI for repair. Contact [email protected] for RMA and safe shipping instructions.
Under certain rare conditions, it may be appropriate for the TIM to be opened and serviced on-site. All TIMs share the same case that can be easily opened for service and placed in use for diagnostics (attached to the Test Head) with the cover removed. DO NOT TOUCH any active electronics with the case removed without direct instruction from RI. TIMs can be safely removed from the Test Head with the power on. The software should be in an idle state (TestExec not running and no testplans loaded) and the Fixture should be removed prior to removing any TIMs.
To open/close the case:
Remove the top half of the Cassini TIM. When placed on a table, The RI model/serial number label should be facing up. When loaded on the left side of Test Head, the TIM port on the upper right side, the side to remove is facing you.
- On the bottom latch, unscrew the 2 left screws 10-15 turns but do not remove the screws completely. There are 4 total, do not loosen or remove the inner pair of screws.
- Unscrew the outer 2 right screws 10-15 turns but do not remove the screws completely. There are 4 total, do not loosen or remove the inner pair of screws.
- Orient the TIM so the text on the TIM label is upright. Unscrew the 8 side screws on both sides. (16 total)
- Pull the bottom latch until the cover clears the EMI/RFI groove of end cover. Pull the cover up and away from the TIM. Note that some TIMs have EMI/RFI elastomer gasket in the groove. Use caution to ensure the elastomer does not come out.
ONLY If cables on the interface block need to be replaced. Unscrew the top 2 screws, then the 2 screws on the side.