This item will not be restocked when depleted and should be avoided for new designs.
This product entered limited availablity on 07/20/2016. View RIKs with Limited Availability for a complete list.
Suggested Replacement: RI8595B
Expiration Reason: Rev B provides higher frequency range at a cost of lower max power
Please visit the Instruments page for the latest Briefs.Browse Instruments »
- 0.7-2.7 GHz Power Amplifier
- 30 Watt Nominal
High Power TIM
The RI8595A provides calibrated power amplifier for high power DUTs.
This is a perfect fit for testing high power DUTs.
TIM Level Performance Data
The RI8595A Test Instrument Module (TIM) provides high power output from 700 MHz to 2.7 GHz. Over most of this range, P1dB exceeds 30 watts. Calibrated output power provides +/-1 dB power accuracy.
The RI8595A can drive into an open circuit without damage.
Frequency Range 700 MHz to 2.7 GHz
Output P1dB, typical
700 MHz to 2.5 GHz 30 Watts (44.7 dBm)
2.5 GHz to 2.7 GHz 10 Watts (40 dBm)
Third Order Intercept, typical 52 dBm
Gain, typical 47 dB
Calibrated Power Accuracy, typical1 ±1 dB
Single TIM Unit
Power Consumption: <100 Watts
TIM Block Type: RF Block (16 Coax)
Related Documents:User account required Create new account