This item will not be restocked when depleted and should be avoided for new designs.
This product entered limited availablity on 01/05/2016. View RIKs with Limited Availability for a complete list.
Suggested Replacement: RIK0267B
Expiration Reason: Placeholder. No rev A was made/shipped.
Aditional replacements may be found by searching for RIK0267 (change the REV letter) or Parallel Handler POD + 15' RIFL Cable.
Allows Parallel communications with Handler (drivers may be necessary). Includes RIFL 2 to RIFL 3 cable (RI Part # GGFEH22B) but NOT Parallel (D-Sub) cable (supplied by customer or MFSHLX1A).
Serial or Parallel generally have the most reliable and fastest interface to handlers versus GPIB and therefore recommended by RI over GPIB interface.
This interface type is the fastest (in reaction time) and preferred method for connecting to a Handler or Prober. The custom cable can be easily completed by adding resistors or bridging the correct pins. Depending on how the handler is configured, a 10K ohm resistor should be installed between each SOT (Start Of Test) pin and pin 13 (+5V for pull up) or pin 12 (ground for pull down).
Handler POD is configured at the factory based on customer provided Handler/Prober interface documentation.
|MFSHLX1A||Handler Cable, 14' DSUB 25pin Pigtail w/ Connector||Create custom cable for Parallel Handler DSUB 25|
Related Documents:User account required Create new account
|Cassini Handler Control Pod Configuration||Manufacturing||Handler Pod,Procedures||Today|
|Site Preparation Guidelines for CASSINI 16 and SPYDER||Admin,Specs||Specifications,Technical Articles & Information,Preparation,Footprint||10/06|
|RI Operating Environment Guidelines||Specs||Specifications,Technical Articles & Information||02/03/2011|
|How to Configure Handler Pod (GPIB/Serial/Parallel)||Manufacturing||Handler Pod,Procedures||Today|
|Handler Control Pod Configuration||Manufacturing||Handler,Pod,Procedures||08/08/2006|
|Handler Control Pod Pinouts|
|Handler Pod Programming and Interface Example (MT9320)|
|Handler Configuration and Interface Cable - Rasco SO1000|
|Item||Ref||Part Number||QTY||Description||Category||Qty in Stock|
|1||GGFEH22B||1||Cable RIFL3-RIFL2 180 in metal collet*||CblMsc|
|2||RI8517A0||1||Handler Pod, MultiSite Parallel RI8517A0*||RInstrmn|